1-Clear the actual calibration
= display the task bar
Rename the calibration file C:\Program Files\AGT\LayoutMaster\Build\CurrentConfigs\LayoutMaster\Options\CalibrationModel.xml by adding _YYYYMMDDHHMM
Reset the Christie Twist to 0.
Open Christie Twist.
Create a new 7x7 grid.
Restart LayoutMaster :
Reset the horizontal and vertical lens shift to 0
2-Adjust the mechanical lens shift
Remove the lens protector or ensure that the lens movements is not obstructed.
Use Edge or Chrome to open the control page of the projector.
User = Admin
Password = Admin
Unlock all lens motors.
Execute the Lens Calibration.
Set the zoom to produce the larger possible image.
Adjust the lens shift so that the image centered on the projecteur center line.
Partially execute the procedure : Master calibration
Level the Master
Make sure the lens shifts are at 0.
Only adjust the focal length and the mechanical lens shifts.
Target a maximum average error of 3mm.
The result will not be perfect at this step but enough to minimize the corrections during the calibration of the grid.
3-Calibrate the projector grid.
Execute the procedure : Grid calibration
4-Calibrate the Master
Fully exucture the procedure Master Calibration without midify the mechanical lens shift.
The final error should be less than the thickness of a line.
If any lens shifts is more than 25, reset the lens shifts to 0, adjust the mechanical lens shift and redo the Grid calibration.
5-Calibrate the lasers
Execute the procedure : Lasers calibration
6-Calibrate the positions
Execute the procedure : Positions calibration